The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||18 March 2010|
|PDF File Size:||5.61 Mb|
|ePub File Size:||1.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Device Manager dialog box appears. The HDD revision can be checked using the following procedure: Push the button under the lock and remove the hard disk.
Click Load to start the calibration measurement. Select the C drive to restore, then click Next button.
Help File – User Manual for E5071C ENA Network Analyzer (Operation and Programming)
A chassis terminal; a connection manul the instrument’s chassis, which includes all exposed metal structure. In this case, follow the instruction to return to the previous registration screen. A message dialog box appears after the completion of the Recovery Image Setup procedure.
Touching the four locations described above with your finger automatically concludes the touch screen calibration. You need to execute initial registration again. If an isolation calibration must be performed using a LOAD standard, nanual the procedure below.
Press the standby switch of the EC to turn it on. The volume label for C: CN or below Returns the contents of the C and D drives to a user-specified state. Safety Safety Summary The following general safety precautions must be observed during all phases of operation, service, and repair of this instrument.
Select a test port and corresponding S e5071cc on which 1-port calibration will be performed. E Figure 4 Item 4: The system recovery procedure is mnual in accordance with the HDD revision. The following screen appears during backup. Inset the key in the lock and turn crockwise. The recovery takes a few minutes depending on the amount of data. If the name label of the calibration kit has been changed, the label is displayed as the soft key.
EC firmware is executed, then the measurement display appears. If you use a calibration kit other than a predefined one, you need to define it.
You cannot use the front panel keys e5071x the initial registration of the EC therefore connect the mouse and the keyboard before turning on the power. If you press F11 twice or more and a screen which is not mentioned in this manual appears, reboot the instrument and restart the process from the first.
The procedure consists of the following items. Select the saved backup image file, then click Next button. Click Select Ports, then select the test ports on which you will perform full 2-port calibration. In EC, right click My Computer. Insert the hard disk into the slot. Turn the screw in anti-clockwise direction until it is locked.
Figure 3 Item 3: This calibration effectively eliminates the frequency response reflection tracking error, directivity error, and source match error from the test setup in a reflection test using that port.
It’s a community-based project which helps to repair anything. Inset the key in the lock and turn anti-clockwise. The System Properties dialog box appears. Right click and select Unistall for both the COM1 ports.
Click OK to proceed. Under certain conditions, dangerous voltage levels may remain even after the power cable has been disconnected. DO NOT Service or Adjust the Instrument Alone – Manuall not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, is present. If anything is missing, contact the nearest Agilent Technologies office.
Close all other applications you currently open. If the connector type of the standard of the calibration kit you use has polarity the distinction between male and femaleyou need to change the standard class definition of the calibration kit depending on the standard you actually use.
Turns on the EC. CN or below This section describes how to create the user backup image. The firmware revision you installed remains even if you perform factory recovery. Minor corrections and updates that are incorporated at reprint do not cause the date to e5071v.
CN or above Step 1. If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered. After restart, the screen for initial registration appears. The dialog box disappears automatically after a few seconds. Test and Measurement Call Center tel 1 fax 1 China: Keep Away from Live Circuits – Operators must not remove instrument covers. Component replacement and internal adjustments must be made by qualified maintenance personnel.
To avoid injuries, always disconnect the power and discharge circuits before touching them.